Efficiency and safety of using a peritoneal dialysis catheter weighted with a stainless steel ballast : the Limousin experience
Objective: To assess the recurrence of PD catheter migration after the introduction of a walnut ballast. Materials and Methods: Retrospective study from 1999 to 2014 of PD patients followed in Limousin. Were compared two groups: ballast group (patients who benefited from the establishment of stainless steel ballast at the intraperitoneal catheter extremity) with 26 patients and control group with 204 patients. The primary endpoint was the occurrence of an episode catheter’s migration after ballast’s establishment. Secondary objectives were (i) to determine the causal factors leading to the catheter weighting, (ii) to ensure the safety of the procedure on the following criteria: infectious complications, mechanicals complications, epurations criteria, and catheter’s survival. Results: More than one year after the implementation of the ballast, no recurrent migration was observed in 86.6% of cases. It wasn’t found an increased risk of infections (OR = 0.5, 95% CI [0.22, 1.13]) or mechanical complications (OR = 1.77- 95% CI [0.77, 4.05]) between the two groups. The adequation criteria were similar: KT / V total : 2.37 in the control group and 2.28 in the ballast group (p = 0.63). The survival of the ballast catheter was comparable among the two groups (p = 0.983). Three causal factors that led to the ballast were identified: automated peritoneal dialysis (APD) (OR = 0.38, 95% CI [0.16, 0.9]), the failure from the first use of the catheter (OR = 19.48, CI 95 % [7.67, 49.48]) and the incarceration of the omentum (OR = 15.84, 95% CI [5.81, 43.21]). Conclusion: The ballast used in these study appears to prevent recurrence of migration, without any impact in terms of infectious or mechanical complications, or on the dialysis criteria or on catheter’s survival. However this catheter does currently not have an EC authorization
T. Dantoine, D. Bénévent, C. Lagarde,C. Leroux-Robert. Lestage du cathéter de dialyse péritonéale. Bulletin Dialyse Péritonéale 1999;9(2).
Cavagna R, Tessarin C, Tarroni G, Casol D, De Silvestro L, Fabbian F. The self-locating catheter: clinical evaluation and comparison with the Tenckhoff catheter. Perit Dial Int J Int Soc Perit Dial. déc 1999;19(6):540‑543.
Di Paolo N, Capotondo L, Sansoni E, Romolini V, Simola M, Gaggiotti E, et al. The self-locating catheter: clinical experience and follow-up. Perit Dial Int J Int Soc Perit Dial. août 2004;24(4):359‑364.
Minguela I, Lanuza M, Ruiz de Gauna R, Rodado R, Alegría S, Andreu AJ, et al. Lower malfunction rate with self-locating catheters. Perit Dial Int J Int Soc Perit Dial. 2001;21 Suppl 3:S209‑212.
Bergamin B, Senn O, Corsenca A, Dutkowski P, Weber M, Wüthrich RP, et al. Finding the right position: a three-year, single-center experience with the « self-locating » catheter. Perit Dial Int J Int Soc Perit Dial. oct 2010;30(5):519‑523.
Chen W-M, Cheng C-L. A simple method to prevent peritoneal dialysis catheter tip migration. Perit Dial Int J Int Soc Perit Dial. oct 2007;27(5):554‑556.
Stucky C-CH, Mason MC, Madura JA, Harold KL. A Technique to Address Peritoneal Dialysis Catheter Malfunction. JSLS. 2010;14(4):608‑610.
Julian TB, Ribeiro U, Bruns F, Fraley D. Malfunctioning peritoneal dialysis catheter repaired by laparoscopic surgery. Perit Dial Int J Int Soc Perit Dial. déc 1995;15(8):363‑366.
Simons ME, Pron G, Voros M, Vanderburgh LC, Rao PS, Oreopoulos DG. Fluoroscopically-guided manipulation of malfunctioning peritoneal dialysis catheters. Perit Dial Int J Int Soc Perit Dial. déc 1999;19(6):544‑549.
Xie J, Kiryluk K, Ren H, Zhu P, Huang X, Shen P, et al. Coiled versus straight peritoneal dialysis catheters: a randomized controlled trial and meta-analysis. Am J Kidney Dis Off J Natl Kidney Found. déc 2011;58(6):946‑955.
Chen G, Wang P, Liu H, Zhou L, Cheng M, Liu Y, et al. Greater omentum folding in the open surgical placement of peritoneal dialysis catheters: a randomized controlled study and systemic review. Nephrol Dial Transplant. 3 janv 2014;29(3):687‑697.
Prischl FC, Wallner M, Kalchmair H, Povacz F, Kramar R. Initial subcutaneous embedding of the peritoneal dialysis catheter--a critical appraisal of this new implantation technique. Nephrol Dial Transplant Off Publ Eur Dial Transpl Assoc - Eur Ren Assoc. août 1997;12(8):1661‑1667.
Crabtree JH, Burchette RJ. Peritoneal dialysis catheter embedment: surgical considerations, expectations, and complications. Am J Surg. oct 2013;206(4):464‑471.
Elhassan E, McNair B, Quinn M, Teitelbaum I. Prolonged duration of peritoneal dialysis catheter embedment does not lower the catheter success rate. Perit Dial Int J Int Soc Perit Dial. oct 2011;31(5):558‑564.
Brown PA, McCormick BB, Knoll G, Su Y, Doucette S, Fergusson D, et al. Complications and catheter survival with prolonged embedding of peritoneal dialysis catheters. Nephrol Dial Transplant Off Publ Eur Dial Transpl Assoc - Eur Ren Assoc. juill 2008;23(7):2299‑2303.
Martínez-Mier G, Luna-Castillo M, Ortiz-Enríquez JJ, Avila-Pardo SF, Fernández V, Méndez-López MT, et al. Factors associated with early peritoneal dialysis catheter replacement in Veracruz, Mexico. Nefrol Publ Of Soc Esp Nefrol. 14 mai 2012;32(3):353‑358.
Outcomes of the Weighted Peritoneal Dialysis Catheter in Patients at Risk of Percutaneous Catheter Failure, par Stephen Stonelake et all
Perit Dial Int March-April 2019 39:142-146; published ahead of print November 25, 2018
Copyright (c) 2019 lariviere benedicte, Rémi Boudet, Marie Essig, Stéphane Bouvier, ali Abdeh, Christine Beauchamp, Monica Ciobotaru
This work is licensed under a Creative Commons Attribution 4.0 International License.